|Color:||Black And Orange||Weight:||Approx 176g|
|Packing:||Color Box||Hold:||Data Hold|
|Product Size:||174*66*30mm||Operation Way:||Auto Range|
|Clamp Size:||32 Mm||Battery:||1.5V*2|
Pocket Clamp Digital Multimeter,
4000 Counts Clamp Digital Multimeter,
4000 Counts clamp on multimeter
4000 Counts Pocket Clamp Digital Multimeter with NCV Flashlight Lighting
VICTOR 606E 4000 Counts Pocket Clamp Digital Multimeter 3 3/4 AC DC Digital Clamp Meter Multimeter Pinza Amperimetrica True RMS
1. SAFETY OPERATION RULE
The meter’s design is in accordance with the CE certification, meets the standard ofIEC61010-1, IEC61010-2-032, IEC 61010-031, IEC61010 CAT.III 600V and Grade II pollution. Read the following before operation.
1.1 Use the meter in accordance with the standards or requirements of this manual, otherwise the protection function provided by the instrument may be damaged or declined.
1.2 To avoid electric shock, be careful when measuring the voltage higher than 60V DC RMS, 30V AC
1.3 Do not input the voltage over the limit value between the terminals or between each terminal and ground.
1.4 Check whether the meter works properly by measuring the voltage or current which were known beforehand. Do not use any more if it is improper or damaged.
1.5 Please check the meter before operation, if the shell is cracked or the plastic parts are damaged, please stop using it again.
1.6 Before operation, please check the test leads, if there are cracks or damage on it, please replace the test leads with same model or the same electrical specifications.
1.7 Do not measure over range of the meter, accessories as probe, and CAT Level.
1.8 When inserting the test leads in the input jack, do not measure the current
1.9 Do not operate with the meter independently
1.10 Please comply with local and national safety regulations. Wear personal protective equipment (such as approved rubber gloves, face masks, and flame-retardant clothing etc.) to prevent the injury from electric shock and arc when charged conductors are exposed.
1.11 When the low battery symbol appears, please replace the battery in time to avoid the wrong values
1.12 Do not use the meter in a humid environment with steam etc., or in the air with explosive gas.
1.13 When using the test leads, please hold your finger behind the finger protective equipment of the probe.
1.14 Please connect the neutral wire or ground wire first under measurement, and then connect the live wire. When the measurement ends, please disconnect the live wire first, and then disconnect the neutral wire and ground wire.
1.15 Before opening the shell or battery cover, remove all probes or test leads. Do not use the meter when it is disassembled or the battery cover opened.
|DC Voltage (DCV)||YES||AC Voltage (ACV)||YES|
|DC Current (DCA)||YES||AC Current (ACA)||YES|
|NCV Deteation||YES||Diode Test||YES|
|True RMS||YES||DCA Zero||YES|
|Auto off||YES||Double Backlight||YES|
|Flashlight lighting||YES||Data Hold||YES|
|DC Voltage: 400mV/4V/40V/400V/600V;±(0.5%+5d)|
|AC Voltage: 4V/40V/400V/600V;±(1.0%+5d)|
|DC Current: 40A/400A/600A;50~60Hz±(2.5%+5d)|
|AC Current: 40A/400A/600A;50~60Hz±(2.5%+5d);Other:±(3.0%+10d)|
|Display Counts: 4000 counts|
|Jaw Size: 26mm|
|Power: 2 x 1.5V AAA|
|Product Size: 200*75*35mm|
|Product Weight: 150g|
Contact Person: sales
Tel: +86 13924679853