|Product Name:||Digital Vibration Meter||Packing Details:||50pcs/Carton; Carton Size: 92.5*34*25cm;9KG/Carton|
|Accuracy:||±10%||Material:||Plastic, Metal, Silica Gel|
Pocket Multifunction Environment Meters,
VC63 Multifunction Environment Meters,
1999um Portable Vibration Meter
1999um Pocket Multifunction Environment Meters Portable Vibration Meter
VC63 Vibration Meter 0~1999um Pocketsize Measure The Displacement, Velocity And Acceleration Of The Objects Electric Power
FUNCTION AND APPLICATION
Vibration meter is a kind of integrated, pocketsize instrument for vibration measurement of rotating machinery. With the capability to measure the displacement, velocity and acceleration of the objects easily, the meter can be applied in industry departments such as electric power, petroleum, chemical engineering and metallurgy to meet different demands.
Vibration measurement range
Displacement: 0~1999um (peak to peak)
Velocity: 0~199.9mm/s (rms)
Linearity and Accuracy
Displacement: 0~20um: ±10% ＞20um: ±5%
Velocity: 0~2.0mm/s: ±10% ＞2.0mm/s: ±5%
Acceleration: 0~2.0 m/s2: ±10% ＞2.0 m/s2: ±5%
Display: 3 1/2 digit LCD
Low voltage indication: If battery voltage ＜5.5V, a battery mark appears on the LCD, then replace the battery with new one.
Hold function: the displayed vibration value is hold and kept for one minute after the MEAS button is released
Power: 6F22 batter (9V)
Battery life: approx. 25h under continuous use
Atmosphere pressure: 75~106kpa
Dimension: 185(H)× 68(W) ×30(D) mm
Weight: 230g (including battery)
Open the battery compartment lid, load a 6F22 battery in the battery compartment with correct polarity, and close the lid.
Checking battery voltage
Press the MEA button and observe the display. If a battery mark appears on the display (left side), replace the battery with new one. If not, the meter is ready for measuring.
Select the acceleration, velocity or displacement measurement mode with the mode selector. The arrow on the right of the display indicates the selected mode.
Force to press the attachment (probe) at right angle to the object surface.
Press the MEAS button (power on) and keep the button depressed, the vibration value is now indicated on the display. By releasing the MEAS button, vibration value at that moment is held on the display. Now you can detach the probe from the object and read the displayed value.
Press MEAS button again to reset the hold function and repeat measurement.
Notice: the attachment should be pressed at right angle to the object surface. If pressed sideways, accurate measurement cannot be made.
This instrument should not be subjected to strong magnetic field, causticity gas and shocks.
Remove the battery when do not to use this instrument for a long time, otherwise electrolyte may leak and cause damage.
Power on failure: battery does not load properly
Abnormal display: the mode selector fail to function or is not on the right position
Contact Person: sales
Tel: +86 13924679853