|Size:||132 Mm X76 Mm X42mm||Product Name:||Analog Multimeter|
|Model No.:||VICTOR7001||Product Packing:||Blister And Cardboard|
|Packing Details:||50pcs/Carton; Carton Size:46.4*31.5*51||Color:||Black And Orange|
Analog VICTOR Digital Multimeter,
Lcd Display VICTOR Digital Multimeter,
500V VICTOR Digital Multimeter
Analog VICTOR Digital Multimeter Large Lcd Display 500V 250mA 1MΩ
VICTOR 7001 Analog Multimeter Large Lcd Display 500V 250mA 1MΩ
Adjust the zero calibration device on the cap to “0” before measuring.
If none of the ranges is working, that means the fuse in the meter might have been damaged. Open the cap to replace with a new fuse. Selected voltage/current range shall be higher than the voltage/current value which you measured. When using “OHM” range, do not measure resistance with the circuit powered on. Discharge the capacitor before measuring its leakage current.
Measuring DC or AC voltage:
Connect the meter across both ends of the measured voltage and read the second scale line.
Measuring DC current:
Connect the meter and the measured circuit in series and read the second scale line.
Short-circuits the test bar and adjust the “OHM” zero potentiometer until the pointer is on “0” of the first “Ω” scale line. If the pointer cannot be adjusted to “0”, it means the internal battery is low. Replace the battery and connect the test bar across both ends of the resistor which to be measured. Read the value of pointer on the first “Ω” scale line and multiply by the factor of the range.
Measuring dry battery:
Connect the test bar across both electrode of the dry battery which to be measured. Load current at full scale is 125mA for 1.5V range and 10mA for 9V range. If the pointer is in the “GOOD” arc range of the third scale line, it means the battery level is high. “BAD” arc means the battery level is low and the arc between “GOOD” and “BAD” means the battery level is about to be low:
Put the Range Switch to “AC10V” and read the fourth “dB” scale line. If the measured circuit has DC component, the test bar should be connected with a DC isolating capacitor above 0.1UF/500V in series. When using other “ACV” ranges to measure dB value, convert the value according to the following table.
DC voltage: (DCV) 0-2.5-10-50-250-500V
2000Ω /V 5.0 grade
AC voltage: (ACV) 0-10-50-500V
2000 Ω /V 5.0 grade
DC current: (DCA) 0-0.5-10-250mA
Voltage drop: 0.5 V 5.0 grade
Resistance: (Ω) 0-10kΩ -1MΩ
Center value: 3.6 5.0 grade (arc length)
Size : 132 mm x76 mm x42mm
|Special Function||VICTOR 7001|
|Battery load capacity test||√|
|Operation Way||Manual range|
|Product color||Case: dark gray|
|Product net Weight||300g (including battery)|
|Standard Accessories||Test leads, battery, instruction manual|
|Carton size||132 mm x76 mm x42mm|
|Standard quantity per carton||40pcs|
|Standard Gross Weight||24kg|
Contact Person: sales
Tel: +86 13924679853